Payment & Shipping Terms:
|Nominal Capacity::||100AH||Temperature Control Method::||Smart P.I.D Control|
|Heating Tube Power::||6 Kw||Turn The Blade Material::||SUS# 316 Stainless Steel|
|Temperature Uniformity::||± 2 ℃||Stirring::||Speed 60r / Min|
|Solution::||PH At 11.0±0.1||Power Voltage::||AC380V 50Hz|
thermal shock test chamber,
salt spray test chamber
± 2 ℃ Laptop / Touch Control Lithium Battery Pack Washing Testing Machine 6kw
Lithium-ion battery stack washing up tester, battery lavation tester is suitable for the following batteries or battery packs
1, hand-held electronic products with lithium-ion battery pack
2, may be placed in the pocket to carry or use of portable electronic products with lithium-ion battery pack
3, other portable electronic products can be placed in the pocket to carry the battery pack
4, the installation of non-user replacement type battery / battery pack of hand-held electronic products such as washing test.
1.Solution: Preparing solution with PH at 11.0±0.1(Naoh solution available mass fraction at 0.004%) and heat solution at (45±2)ºC.
2. Soaking: Fix the sample on the rotating device, and then place the sample in the solution (the center of the battery is 300mm + 10mm) to soak 0.5H. The temperature of the solution keeping at (45 + 2) ºC during the test.
3. Stiring: Fix the sample on the rotary device and rotat 0.5H in the solution, rotating speed at 60r/min.
4. Dewatering:Remove the solution, rotate 10 min, and rotating speed is 800r/min.
5. Drying: Put the samples in a high temperature box to heat 0.5H, and the heating temperature was (45 + 2) ºC.
6. If the sample can continue to charge and discharge, then follow the prescribed charge and discharge method to do a cycle of discharge charging and end the test. If the sample can not continue to charge and discharge, the test will be finished. During the test, the sample should not fire and explode.
|Product name||Lithium-ion battery stack washing up tester, battery lavation tester|
room temperature ~ 80 ℃ (adjustable, can be arbitrarily set), the commonly
used temperature of 45 ± 2 ℃
|temperature accuracy||± 0.5 ℃|
|temperature uniformity||± 2 ℃|
the sample fixed to the rotating device (washing paddle) in the solution for
0.5h, speed 60r / min
|inner box Material||SUS # 316 stainless steel|
|test time||0-9999H / M / S can be arbitrarily set|
|temperature control methods||intelligent P.I.D control|
PLC man-machine interface, reliable performance, high modulus accuracy,
set the parameters after the test automatically completed, simple and efficient
configure the pH value of 11.0 ± 0.1 solution (can use the mass fraction of
0.004% NaOH solution), the washing solution by the customer self
the sample fixed on the rotating device (washing paddle), and then placed
the sample in the solution (the battery center distance from the surface height
of 300mm ± 10mm) soak 0.5h, the test process to maintain the temperature
of the solution (45 ± 2) ℃
|equipment outside size||1000x1260x2200 mm (LxWxH)|
|Power supply||AC 380V, 50HZ, 16A (need to be equipped with independent air switch),|
Contact Person: Lily Sun