DONGGUAN YUYANG INSTRUMENT CO.,LTD
China Good Quality Manufacturer of Testing Equipment
|Place of Origin:||China|
|Certification:||UL 1642:2012 ，UL 2054: 2012，GB/T 31241-2014，GB/T 18287 -2013，GB/T 8897.4-2008，YD/T 2344.1-2011，GB/T 21966-2008，UN38.3(2012)，IEC62133-2012，IEC 62281: 2004，MT/T 1051-2007，YD 1268-2003，GB/T 19521.11-2005，YDB 032-2009，IEC 60086: 2007|
|Packaging Details:||Standard Export Wooden Case|
|Delivery Time:||10-15 work days|
|Payment Terms:||L/C, T/T, Western Union|
|Supply Ability:||100 sets/month|
|Maxcapacity:||<1000KN||Burner:||The UL Burner,caliber Diameter 9.5mm Or 38mm, Length 100 Mm|
|Surface Diameter Of Test Hole:||102mm||Flame Application Time::||3.0-999.9±0.1 Seconds|
|Flame From Mesh Screen Height::||8.38mm||Test Octagonal Mesh Height::||12"(305mm)|
thermal shock test chamber,
salt spray test chamber
UL1642 Battery Safety Tester Projectile Burning Testing Equipment
Battery Burning Tester is suitable for lithium battery , battery pack, mobile phone, notebook computer and other small electronic consumer products battery fire resistance/flammability testing. The main function is to observe a series changes and the impact force size produced by exploding after being heated by flames.
1. Place the test lithium battery or battery pack on a wire mesh, the wire mesh is located on a circular hole with a diameter of 102mm(4 in.). The wire mesh consists of 20 mesh wires per inch with a diameter of 0.017 inches(0.43mm)
2. The wore mesh is placed about 38mm above flame.
3. An octagonal aluminum wire mesh is covered around the sample.The specification of the aluminum wire mesh is as follows: parallel to each other on both sides of distance(610±10)mm, height(305±5)mm.Aluminum wire mesh specifications for aluminum wire diameter 0.25mm,mesh number 16~ 18.
4. Make sure the test specimen is heated on the wire until it explodes or the lithium battery is ignited and burned up.
Test results: after the test, components of the battery(except dust products) or the battery as a whole shall not penetrate the aluminum mesh.
UL 1642:2012 ,UL 2054: 2012,GB/T 31241-2014,GB/T 18287 -2013,GB/T 8897.4-2008,YD/T 2344.1-2011,GB/T 21966-2008,UN38.3(2012),IEC62133-2012,IEC 62281: 2004,MT/T 1051-2007,YD 1268-2003,GB/T 19521.11-2005,YDB 032-2009,IEC 60086: 2007
|Burner:||bunsen burner,inner diameter of nozzle is 9.5mm and length is about 100mmvv|
|Flame application time:||0-999.9±0.1 seconds|
|Surface diameter of test hole:||102mm|
|Test mesh screen:||woven from stainless steel wire with a diameter of 0.43mm(0.017inch) into a mesh screen with 20 meshes.|
|Ignition controller:||remote ignition, automatic control of the test process.|
|Remote controller distance:||no barrier, 10 meters|
|Flame from mesh screen height:||38mm|
|Test octagonal mesh width:||the distance between each opposite is 24"(610mm),octagonal, and the octagonal is movable for easy replacement and removal.|
|Test octagonal mesh height:||12"(305mm)|
|Octagon cover material:||aluminum mesh,consisting of aluminum wire with a diameter of 0.010inch(0.25mm) woven into a net containing 16-18 grids per inch.|
|Test part:||stainless steel material, one side of which is 5mm thick tempered glass,which can witness the test status.Meanwhile, the top of the protective box has exhaust fans, and an external exhaust pipe interface is reserved.|
|Test shelter volume:||750*750*1000mm|
|Combustion gas:||methane(Customer should bring along their own)|
3.Conbustion gas:liquefied petroleum gas or industrial methane(customer should bring along their own)
Contact Person: Lily Sun