DONGGUAN YUYANG INSTRUMENT CO.,LTD
China Good Quality Manufacturer of Testing Equipment
|Place of Origin:||China|
|Certification:||GB4706, IEC60335, UL859|
|Packaging Details:||Standard Export Wooden Case|
|Delivery Time:||5-8 work days|
|Supply Ability:||30 Set/Sets per Month|
|The Test Rate::||10 To 30 Times Per Min||Power Supply:||220V 50Hz|
|Number Of Flex Per Test:||1-999999 (selectable)||Rate Of Flexing:||10~60 Times/min(selectable)|
|Flexing Angle:||Swivel Angle On Either Side For 45° , 60°, Or 90°(or 180°)(selectable)||Weight:||10N/20N/133g/284g/500g*6(selectable)|
cable testing equipment,
spark testing equipment
10~60 times/min Metal Wire And Cable Repeated Bending Testing Machine
UL817 IEC60335 cable bending testing machine can test six cables or cord sfitted with plugs simultaneously. This improves testing capacity.IEC60884-1 calls for 10,000 flexing at a rate of 60 flexing per minute. Halfway through the test, the cords need to be rotated 90°. The plug sockets are designed to make this rotation easy and accurate.
This tester has individual station counters so that any failure data can be retrieved by station.The cables are connected to a socket and the tail of the cable is connected to the tester to detect if an internal fault occurs with the cable wiring or accessory connection.
The instrument starts in a vertical orientation and then flexes the specimens 45° on either side of the vertical for a total 90° of flexing. A standard rate of 60 flexing per minute is used but a slower rate can be selected if desired.
A set of customer selected weights are included with the instrument so that the proper force can be applied to the cable during the test.
|Power Supply||220V 50Hz|
|Number of Flex Per Test||1-999999 (selectable)|
|Number of Test Station||6,lndividual counts for each station|
|Rate of Flexing||10~60 times/min(selectable)|
|Flexing Angle||Swivel angle on either side for 45° , 60°, or 90°(or 180°)(selectable)|
|Dimensions||W*D*H=100cm *50cm* 112cm|
|Conforms to standard||IEC884-1 Figure 21 and clause 23.4,IEC320-1,VDE0620|
Contact Person: Cc Dai