|Place of Origin:||China|
|Certification:||EN 50281-2-1, ASTM E1491-06, ISO/IEC 80079-20-2, GB/T 16429|
|Minimum Order Quantity:||1 set|
|Packaging Details:||Plywood Box|
|Delivery Time:||25 work days|
|Payment Terms:||T/T, L/C, Western Union|
|Supply Ability:||4 sets per quarter|
|Product Name:||Flammability Testing Equipment||Usage:||Minimum Ignition Temperature Tester|
|Standard:||EN 50281-2-1, ASTM E1491-06, ISO/IEC 80079-20-2, GB/T 16429||Furnace Operating Temperature Range:||Room Temperature ~ 800 ℃|
|The Pressure Of The Dispersion Air Chamber:||0 ~ 1 Bar||Power:||230V, 50HZ|
|Size:||500X300X450 Mm||Temperature Accuracy:||± 1% (higher Than 500 ºC), ± 3% (lower Than 300 ºC)|
flame test equipment,
flammability test apparatus
EN 50281-2-1 / IEC 80079-20-2 Combustible Dust Minimum Ignition Temperature Tester
This machine is a way of testing the sensitivity of the ignition that the dust cloud is in the heated environment. When a lot of dust is exposed in heating the air and the air temperature is high enough, it may lead to spontaneous combustion. This method can test the minimum ignition temperature of combustible dust cloud.
EN 50281-2-1: Methods for determining the minimum ignition temperatures of dust.
Method B: Dust cloud in a furnace at a constant temperature.
ASTM E1491-06: Standard test method for minimum auto ignition temperature of dust clouds.
IEC 61241-2 replaced by ISO/IEC 80079-20-2:2016.
Electrical apparatus for use in the presence of combustible dust. Test methods: Methods for determining the minimum ignition temperatures of dust.
ISO/IEC 80079-20-2:2016 Explosive atmospheres. Material characteristics. Combustible dusts test methods
GB / T 16429 "minimum ignition temperature of a dust cloud testing methods"
1. The Test is carried on the in Godbert-Greenwald furnace.
2. The heating furnace is consisting of the vertical quartz tubes. The hole is at the bottom of the tube.
3. Furnace operating temperature range: room temperature ~ 800 ºC
4.The operating temperature can rise to 1000 ºC to clean the test chamber.
5. Temperature accuracy: ± 1% (higher than 500 ºC), ± 3% (lower than 300 ºC).
6. The pressure of the dispersion air chamber: 0 ~ 1 bar
7. The pressure accuracy of the air storage device: ± 5%
8. Size: 500X300X450 mm.
9. Power: 230v / 50Hz or 110v / 60Hz.
The Test is carried on the in Godbert-Greenwald furnace. The heating furnace is consist of the vertical quartz tubes. The hole is at the bottom of the tube, the glass observing room is connected with sample fixing device at the top of the quartz tubes. The compressed air from the tank blows the dust into the heating furnace and the sample fixing device, and then exposed into the glass observing room and the quartz tubes .the heating furnace is located on the holder . And you can observe the ignition appearance from the bottom of the furnace. The test begins from the 500ºC. If you observe the ignition appearance, you shall decrease the temperature at the 10ºC, until the ignition does not appear.
When the ignition occurs, the lowest temperature is recorded as the minimum ignition temperature of the dust cloud.
The minimum ignition temperature of the test data information is used to determine whether the surface temperature of the equipment can cause the automatic fire of the exposed dust . It is an important standard to evaluate the selection of the equipment at the condition of the dusty environment. When using the minimum ignition temperature testing data ,you shall leave a safety limit for the small-scale test .
Contact Person: Seline Yu