DONGGUAN YUYANG INSTRUMENT CO.,LTD
China Good Quality Manufacturer of Testing Equipment
|Place of Origin:||China|
|Certification:||EN ISO 9239-1, ASTM E648, ASTM E970, NFPA 253|
|Minimum Order Quantity:||1 set|
|Packaging Details:||Plywood Box|
|Delivery Time:||20 work days|
|Payment Terms:||T/T, L/C, Western Union|
|Supply Ability:||2 sets per month|
|Product Name:||Flammability Testing Equipment||Usage:||Flooring Radiant Panel Test Apparatus|
|Material:||Stainless Steel||Heat Flux Measurement Range:||(0-15) Kw / M2|
|Exhaust Speed:||(2.5 ± 0.2) M / S||Exhaust Capacity:||(39-85) M3 / Min|
|Gas Flow Rate:||(0.026 ± 0.002) L / S||Flame Height:||60 ~ 120mm|
flammability test apparatus,
flammability test chamber
Flooring Radiant Panel Test Apparatus, 0~15KW/W2 Heat Flux Flammability Testing Equipment
This test method is used to measure the critical radiant flux of horizontally-mounted floor covering systems exposed to a flaming ignition source in a graded radiant heat environment, within a test chamber. It can also be used to measure this same critical radiant flux for exposed attic floor cellulose insulation.
The European Union is using this test method for fire classificaiton of flooring products, throughout European Member States.
The radiant heat is applied by means of a gas-fuelled panel, inclined at 30º, and directed at a horizontally mounted floor covering system specimen. The radiant panel generates a radiant energy flux distribution ranging from a nominal maximum of 10.9 kW/m2 to a minimum of 1.1 kW/m2.
A small stainless steel pilot burner assists in specimen ignition.
The distance burned until flame-out is reached and converted, by calibration, into an equivalent critical radiant flux, in kW/m2.
A smoke measuring system, according to DIN 50055, is mounted on a separate frame at the exhaust stack.
Traditionally, tests are conducted to give a maximum critical radiant heat flux of 10.9 kW/m2, but it can optionally be conducted with higher radiant heat input, of up to 25.0 kW/m2.
1. Angled Control Rack for convenience in use, allowing observation of the apparatus and controls duing equipment set-up and calibration.
2. Automatic ignition of the radiant panel and safety cut-out.
3. Hinged access to test area, via 2 position door with observation window and sliding platform.
4. Stainless steel hood with smoke measurement ports.
5. Data acquisition and analysis software
Contact Person: Jackson Sun